TG-51 Calibration of First Commercial MRI-Guided IMRT System in the Presence of 0.35 Tesla Magnetic Field
S Goddu*, O Pechenaya Green, S Mutic, Washington University School of Medicine, Saint Louis, MOWE-G-BRB-8 Wednesday 4:30:00 PM - 6:00:00 PM Room: Ballroom B
The first real-time-MRI-guided radiotherapy system has been installed in a clinic and it is being evaluated. Presence of magnetic field (MF) during radiation output calibration may have implications on ionization measurements and there is a possibility that standard calibration protocols may not be suitable for dose measurements for such devices. In this study, we evaluated whether a standard calibration protocol (AAPM-TG-51) is appropriate for absolute dose measurement in presence of MF.
Treatment delivery of the ViewRay (VR) system is via three 15,000Ci Cobalt-60 heads positioned 120-degrees apart and all calibration measurements were done in the presence of 0.35T MF. Two ADCL-calibrated ionization-chambers (Exradin A12, A16) were used for TG-51 calibration. Chambers were positioned at 5-cm depth, (SSD=105cm: VR's isocenter), and the MLC leaves were shaped to a 10.5cm x 10.5 cm field size. Percent-depth-dose (PDD) measurements were performed for 5 and 10 cm depths. Individual output of each head was measured using the AAPM-TG51 protocol. Calibration accuracy for each head was subsequently verified by Radiological Physics Center (RPC) TLD measurements.
Measured ion-recombination (Pion) and polarity (Ppol) correction factors were less-than 1.002 and 1.006, respectively. Measured PDDs agreed with BJR-25 within ±0.2%. Maximum dose rates for the reference field size at VR's isocenter for heads 1, 2 and 3 were 1.445±0.005, 1.446±0.107, 1.431±0.006 Gy/minute, respectively. Our calibrations agreed with RPC-TLD measurements within ±1.3%, ±2.6% and ±2.0% for treatment-heads 1, 2 and 3, respectively. At the time of calibration, mean activity of the Co-60 sources was 10,800Ci±0.1%.
This study shows that the TG-51 calibration is feasible in the presence of 0.35T MF and the measurement agreement is within the range of results obtainable for conventional treatment machines.